..::::..

# Capacitance Meter

This project is more complex than the others described earlier. However, when finished, you will have an instrument capable of measuring all but the largest capacitors used in radio circuits. Unlike variable resistors, most variable capacitors are not marked with their values. As well, the markings of capacitors from salvaged equipment often rub off. By being able to measure these unmarked components, this project will prove useful to the constructor, vintage radio enthusiast or antenna experimenter.
The common 555 timer IC forms the heart of the circuit (Figure Three). Its function is to charge the unknown capacitor (Cx) to a fixed voltage. The capacitor is then discharged into the meter circuit. The meter measures the current being drawn through the 47 ohm resistor. The 555 repeats the process several times a second, so that the meter needle remains steady.
The deflection on the meter is directly proportional to the value of the unknown capacitor. This means that the scale is linear, like the voltage and current ranges on an analogue multimeter.
The meter has five ranges, from 100pF to 1uF, selected by a five position two pole switch. In addition, there is a x10 switch for measuring higher values and a divide-by-two facility to allow a better indication on the meter where the capacitor being measured is just above 100, 1000pF, 0.01, 0.1 or 1 uF.
Component values are critical. For best accuracy, it is desirable that the nine resistors wired to the Range switch have a 2% tolerance. If 0A47 diodes are not available, try OA91 or OA95 germanium diodes instead. Construct the meter in a plastic box; one that is about the size of your multimeter but deeper is ideal. The meter movement should as large as your budget allows; you will be using it to indicate exact values. A round 70mm-diameter movement salvaged from a piece of electronic equipment was used in the prototype. The meter you buy will have a scale of 0 to 50 microamps. This scale needs to be converted to read 0 to 100 (ie 20, 40, 60, 80, 100 instead of 10, 20, 30, 40, 50). Use of white correction fluid or small pieces of paper will help here.
The components can be mounted on a piece of matrix board or printed circuit board. Use a socket for the IC should replacement ever be needed. Keep wires short to minimise stray capacitance; stray capacitance reduces accuracy.
Calibrating the completed meter can be done in conjunction with a ready-built capacitance meter. Failing this, a selection of capacitors of known value, as measured on a laboratory meter, could be used. If neither of these options are available, simply buy several capacitors of the same value and use the one which is nearest the average as your standard reference. Use several standards to verify accuracy on all ranges.
To calibrate, disable both the x10 and divide-by-two functions (ie both switches open). Then connect one of your reference capacitors and switch to an appropriate range. Vary the setting of the 47k trimpot until the meter is reading the exact value of the capacitor. Then switch in the divide-by-two function. This should change the reading on the meter. Adjust the 10k trimpot so that the needle shows exactly twice the original reading. For example, if you used a 0.01 uF reference, and the meter read 10 on the 0.1 uF range, it should now read 20. Now switch out the divide-by-two function.
If you are not doing so already, change to a reference with a value equal to one of the ranges (eg 1000pF, 0.01uF, 0.1uF etc). Switch to the range equal to that value (ie the meter reads full-scale (100) when that capacitor is being measured. Switching in the x10 function should cause the meter indication to drop significantly. Adjust the 470 ohm trimpot so that the meter reads 10. Move down one range (eg from 0.01uF to 1000pF). The meter should read 100 again. If it does not, vary the 470 ohm trimpot until it does. That completes the calibration of the capacitance meter. Now try measuring other components to confirm that the measurements are reasonable.
With care, an accuracy of five percent or better should be possible on most ranges.

Source: Electronics Lab

 ..:: Important ::.. Read disclamer, please!
 ..:: Diagrams ::.. Audio Controllers and timers For your car Light and LED Miscellaneous PC related Power supplies Radio Telephone related Tools and measuring
 ..:: Newest 10 ::.. Simple Adjustable Voltage Source Telephone In-Use Relay Controller Astable Multivibrator LM317T Voltage Regulator with Pass Transistor LM317T Variable Voltage Regulator Decibel Meter Touch Activated Light Low Frequency Sinewave Generators LED Photo Sensor Telephone Audio Interface
 ..:: Top 10 ::.. VGA to TV Scart 2N3055 Power Amplifier 60W LM317T Variable Voltage Regulator 12 Volt 30 Amp PSU 6 Channel Disco Running Lights Transformerless Power Supply Adjustable Strobe Light Class-A-AB Amplifier 100w 6 x 6 Loop Antenna Digital Radar Speedometer

Cialis 5 mg senza ricetta Effetti collaterali del Cialis 5 mg Ordina Cialis 20mg Consegna a domicilio di Cialis originale in Svizzera Acquista Viagra in Svizzera senza prescrizione Cialis 10mg senza ricetta Dove comprare Viagra in farmacia online Sconto Cialis 5mg Dónde adquirir Reduslim Spedra Avanafil in vendita online Acquista Viagra in Svizzera senza prescrizione Dónde comprar Cialis Comprar Viagra Contrareembolso Descuento Cialis 5 mg Comprar viagra original Efectos secundarios de Cialis 5 mg Reduslim resultados Remedios para la Disfuncion Erectil Cilis vs Viagra Aviator ???? ?? ?????? Acquisto Cialis 5 mg online Cialis 5 mg effetti collaterali Comprare Cialis 20mg a buon prezzo Consegna a domicilio di Cialis originale in Svizzera Acquisto online di Viagra senza ricetta in Svizzera Cialis 10mg Acquistare Viagra in farmacia online Costo del Cialis 5mg Reduslim opiniones Comprare Spedra Avanafil senza ricetta online Dove comprare il Viagra in Svizzera Comprar Cialis barato precio Viagra genérico Descuento Cialis 5 mg Viagra sin receta Cialis 5 mg efectos secundarios severos Reduslim pérdida de peso Problemas de erección tadalafilo en venta avanafil prezzo cialis 20 mg prezzo cialis effetti benefici cialis 10 mg Reduslim Kaufen pulse x coin price avanafil prezzo buy cialis or generic tadalafil ?????? ?????? ??????????? ?????? ???????? ?????? ?????? ????????? ?????? ?????? ???????????? ?????? ?????? ? ?????? ??????????? http://diplomy-originaly.com ?????? ?????? ????? ?????? ?????? ? ??????? ??????????? ?????? ?????? ???????? ?????? ?????? ???????????? ?????? ?????? ???????????? http://diplomsagroups.com ?????? ?????? ????????? ??? ?????? ?????? ?????? ?????? ????????? ???? ?????? ?????? ? ??????? ??????????? ?????? ???????? http://sreestrom.ru/ ?????? ???????? ?? 11 ????? ?????? ?????? ?????????? ?????? ?????? ???? ?????? ?????? ? ???????? ?????? ?????? ? ?????? ??????????? https://russian-diplom.com ?????? ?????? ????????? ? ?????????? ? ?????? ?????? ?????? ???????????? http://rudiplomirovanie.com ?????? ?????? ? ??????? ?? ? ?????? ?????? ?????? ???? ?????? ?????? ???????? ?????? ?????? ? ??????? ?? ? ?????? ?????? ???????? https://premiums-diploms.com ?????? ?????? ? ???????? ?????? ?????? ????? http://russdiplomiki.com ?????? ?????? ????? ? ?????????? ? ?????? ?????? ?????? ??????????? https://diplomansy.com ?????? ?????? ? ?????????? ? ?????? ?????? ?????? ?????? ???? https://diploms-asx.com ?????? ?????? ? ?????????? ? ?????? ?????? ?????? ?????? ? ??????? ??????????? ?????? ?????? ????????? ???? ?????? ?????? ? ?????????? ? ?????? ???? ?????? ?????? ? ??????? ??????????? https://eonline-diploms.com ?????? ?????? ????????? ? ?????????? ? ?????? ?????? ????????????? ? ???????? http://free-diplommi.com ?????? ????????? ?????? ?????? ?????? ????????????? ? ???????? http://diplomrussian.com ?????? ?????? ?????????? ?????? ?????? ?????? ? ?????? ??????????? ?????? ?????? ???????????? ?????? ?????? ???? ? ?????????? ? ?????? ?????? ?????? ????????? ?????? ?????? ???????? ?????? ?????? ???? ? ?????????? ? ?????? ?????? ?????? ???????? ?????? ?????? ?????? ?????? ? ?????????? ? ?????? ???? ?????? ?????? ???????? ?????? ???????? ?????? ?????? ????????? ? ?????????? ? ?????? ?????? ????????????? ? ????? ?????? ?????? ????? ?????? ?????? ????????? ? ?????????? ? ?????? ?????? ?????? ? ?????? https://diplomsy-landsy24.ru/ ?????? ?????? ???? ? ?????????? ? ?????? ??? ?????? ?????? http://diploms-service.com ?????? ?????? ????????? ? ?????????? ? ?????? ?????? ?????? ???????????? https://kyc-diplom.com ?????? ??????????? ?????? ? ?????????? ? ?????? ?????? ?????? ???????????? ??? ?????? ?????? ?????? ?????? ? ???????? ??? ?????? ???????? https://originaly-dokuments.com ?????? ?????? ? ?????????? ? ?????? ?????? ?????? ?????? ???? https://diplomy-grup24.com/ ?????? ?????? ? ?????????? ? ?????? ????